What causes Hardware error or Instrument error on a DR6000 during a measurement in the UV range?
Hardware error or Instrument error DR6000
In the most cases the UV lamp is damaged and not able to start; if power cycling the instrument does not resolve the issue, the lamp will need to be replaced. Please switch off the DR6000 and replace the UV lamp. For instructions please see the user manual.