Why will my CL10 chlorine sensor not calibrate after changing the electrolyte?
CL10 chlorine sensor cap loose, --.-- reading low
The most common problem with the CL10 not calibrating or reading low is that the membrane cap is not on tight enough. Please turn the cap until it actually stops turning. This brings the membrane closer to the electrode and increase the analytical response.