How is the MODBUS Map created for the FL1500?
MODBUS Map creation for the FL1500
Each data log channel is mapped to a Modbus measurement register, therefore the 16 data log channels are tightly coupled to the Modbus measurement registers. Each Modbus measurement register is updated when the corresponding data log channel is updated. The FL1500 dynamically maps the user selected measurements to the 16 available data log channels. This internal mapping remains static until a new program is sent to the FL1500 with a different selection of measurements to data log.