Are there instructions available for how to use the Sludge Judge Samplers?
Instructions on how to use the sludge judge samplers
The instructions for using the Sampler, Water Core, "Sludge Judge" Standard Capacity, 3/4" Diameter (Product # 2194300) and Sampler, Water Core, "Sludge Judge" Large Capacity, 1-1/4" diameter (Product # 2506102) are as follows:
- Slowly lower the Sludge Judge® to the bottom of the tank. The float valve opens (see figure A), allowing materials to flow in. When the bottom has been reached and the pipe has filled to the surface level, tug slightly on the rope as the unit is raised. This sets the check valve(see figure B), trapping the mixture inside
- When the unit has been raised clear of the liquid, the amount of solids in the sample can be read using the 1 ft. (0.31 m) increments marked on the pipe sections. To release the material in the unit, touch the pin extending from the bottom section against a hard surface. This opens the check valve to drain the sample.