What can cause an LDO probe to read 'Out of Range'?
LDO probe reads Out of Range causes
Water has most likely gotten into the probe or pressure module. If water incursion is suspected, the probe can be placed in a dessicator overnight to see if the moisture can be removed. The pins on the connector should also be checked.