Peaks are close together on the Lachat Ion Chromatography (IC) analyzer. Is there a way to achieve better peak resolution or better separation of analytes?
Lachat IC separation
Yes. In addition to replacing the column(s) one can consider weakening the eluent or slowing the eluent pump flow rate. Some Lachat IC methods will have specific details regarding this.