If the lid is open when the 2100Q and 2100Q ISO turbidimeter is powered on, will this affect the calibration?
Calibration with lid open
2100Q instrument does not perform any zero/dark measurement during startup so this will not be a concern with this instrument. The zero point is set during calibration. The values are stored with the adjustment during calibration. The instrument will not let you calibrate with the lid open.