How clear Calibration Required or Replace Sensor warnings for pH or ORP sensor?
Clear Calibration Required or Replace Sensor warnings for pH or ORP sensors.
There was a software version issue for the Differential pH and ORP sensors that caused the calibration reminder and sensor days counter to not count correctly or be able to be cleared. This was an issue with sensor software version 2.0 and 2.01. If this issue happens, a firmware update to V2.1 or greater will fix the issue on the sensors. This is true for any of the sc controllers.