What is the maximum time out period for a flow pulse signal (contact closure or pulse from a flowmeter) on the AS950 sampler?
AS950 samplers minimum and maximum pulse hold time information.
The minimum pulse hold time for the AS950 is 50 milliseconds. If the pulse is shorter than 50 milliseconds then the sampler will not recognize the pulse signal. The default maximum pulse hold time is 2.0 seconds (longer holds will not trigger a sample).The maximum hold time can be modified in the hardware setup menu to a value between 2 seconds and 300 seconds.