How can slow stabilization time on the CDC401 electrodes be resolved?
CDC401 slow stabilization
Slow stabilization time can be caused by:
- Mineral or sample buildup on electrodes which can be resolved by cleaning the probe (refer to the User Manual for Conductivity Probe Models: CDC40101, CDC40103, CDC40105, CDC40110, CDC40115 or CDC40130).
- Bubbles trapped under the probe tip which can be resolved by gently shaking the probe side to side to dislodge any trapped air bubbles.