Can anything be done when the power does not come on for the SL1000 Portable Parallel Analyzer™ (PPA)?
SL1000 Portable Parallel Analyzer™ (PPA) does not power on
First, make sure the battery has been charged (the unit can turn on during charging, depending on how charged it is). If the battery has been charged try holding down the power button for 20 seconds. If this powers it on, perform a restart through the instrument menu: MAIN MENU -> SETTINGS -> METER -> RESTART .
Additionally, if the version of the software on the SL1000 is earlier than version 184.108.40.2069 running a software update can often resolve this issue.
This information applies to the following:
Full SL1000 Portable Parallel Analyzer™ (PPA) Kit (Product # 8499000)
SL1000 - PPA Portable Parallel Analyzer- Portable Colorimeter with USB with EPA approved Free and Total Chlorine and Copper tests (Product # 9430000)
Full SL1000 Portable Parallel Analyzer (PPA) Nitrification Kit (Product # 25632200)